to reading the User’s Manual. Please read the Accutrend. Plus User’s Manual carefully before using the meter for the first time. The Accutrend Plus meter is used. The Accutrend Plus system, one meter delivering both total cholesterol and glucose test results, can with Accutrend Plus may help employers improve employee productivity, improve .. 15 Cholestech LDX System User Manual. Inverness. View and Download Cobas Accutrend Plus user manual online. Accutrend Plus Blood Glucose Meter pdf manual download.
|Published (Last):||24 February 2018|
|PDF File Size:||6.59 Mb|
|ePub File Size:||20.11 Mb|
|Price:||Free* [*Free Regsitration Required]|
Regularly check that all display elements are functioning correctly to prevent misinterpretations due to a defective display.
Further Information Accutrend Cholesterol Control Control solution to be used accktrend Accutrend Cholesterol test strips Accutrend Plus instrument kit Meter kit Product limitations Please read the insert packaged with the test strips or control solutions for majual information on product data If the measured val- ues are to be stored with time information, enter the correct settings see the Setting the date section of this manual.
Now check whether the displayed result falls within the acceptable range. Target values which should be obtained can either be found on the labels or on the package inserts for the test strips or control solutions.
Press the M button to change the currently flashing setting. Instrument settings Setting the time format Now select the time format to be used: Cleaning Cleaning the outer instrument components Ensure the instrument is powered off. The code number last stored or used is displayed after powering on the instrument.
Always power off the instrument before accturend it! Operating The Instrument Operating the instrument Operating the instrument Before using the instrument for the first time, perform the following steps: Page 90 Quality control recommendation The measurement time varies depending on the accutrenr parameter.
Caution, consult accompanying documents. The instrument must be powered off before you can activate the set mode.
Press the M button to change the currently flashing figure. Performing a measurement Plausibility check after glucose measurement: Page 95 Memory To navigate within one memory area, use the M button. In addition, you accturend flag measurements carried out with control liquids as controls. Quality Control Recommendation Quality control recommendation Quality control recommendation To ensure that the instrument is functioning properly, you should regularly use two levels of control solutions for a function check.
Page 8 Contents Instrument settings Brief overview of the instrument settings Moving nanual is not possible. Each time you accitrend the Set button, the last stored value of each memory area, if any, is displayed. Integrated Control Functions Introduction Integrated control functions The Accutrend Plus instrument has numerous integrated or available control functions, including the following: Performing a measurement Alternative blood application outside the device: Repeat the measurement using a new test strip.
Blood can be applied either in the instrument or outside the instrument. Instrument Settings Operating the instrument Instrument settings Brief overview of the instrument settings The following table provides an overview of the available settings.
Cobas Accutrend Plus User Manual
Consult instructions for use. Page Alphabetical index Date Quality control recommendation After switching on the instrument, check the following displays: If any problems occur during coding, the meter displays an error message see Error Messages chapter of this manual.
Press the Set button to save the selected setting and to continue setting the time format. Performing a measurement Sample material Fresh capillary blood is used as sample material. Quality control recommendation Press the M button to flag the measurement as a control bottle symbol is displayed. Press the Set button to switch to the code number for the other parameter.
Manufactured for and distributed in the U.
COBAS ACCUTREND PLUS USER MANUAL Pdf Download.
Optionally, you can assign one of nine different events to a measurement. Press the M button to show the next oldest measured value within one memory area. Page 81 Quality control recommendation What you need: Close the container immediately after removing strip to protect the desiccant and strips, otherwise the test strips may become unusable before their expiration date.